National Instruments FPT-1015 manual
- Read online or download the manual
- 33 pages
- 0.58 mb
Go to page of
Similar user manuals
National Instruments NI PCI-654132 pages 0.95 mb
National Instruments CFP-222038 pages 4.73 mb
National Instruments PCI-485/2 Isolated19 pages 0.13 mb
National Instruments 372139B-01301 pages 1.77 mb
National Instruments NI 923736 pages 0.51 mb
National Instruments NI PCIe-836x37 pages 0.62 mb
National Instruments Measure Data Acquisition98 pages 2.44 mb
National Instruments PC-LPM-16/PnP125 pages 1.66 mb
A good user manual
The rules should oblige the seller to give the purchaser an operating instrucion of National Instruments FPT-1015, along with an item. The lack of an instruction or false information given to customer shall constitute grounds to apply for a complaint because of nonconformity of goods with the contract. In accordance with the law, a customer can receive an instruction in non-paper form; lately graphic and electronic forms of the manuals, as well as instructional videos have been majorly used. A necessary precondition for this is the unmistakable, legible character of an instruction.
What is an instruction?
The term originates from the Latin word „instructio”, which means organizing. Therefore, in an instruction of National Instruments FPT-1015 one could find a process description. An instruction's purpose is to teach, to ease the start-up and an item's use or performance of certain activities. An instruction is a compilation of information about an item/a service, it is a clue.
Unfortunately, only a few customers devote their time to read an instruction of National Instruments FPT-1015. A good user manual introduces us to a number of additional functionalities of the purchased item, and also helps us to avoid the formation of most of the defects.
What should a perfect user manual contain?
First and foremost, an user manual of National Instruments FPT-1015 should contain:
- informations concerning technical data of National Instruments FPT-1015
- name of the manufacturer and a year of construction of the National Instruments FPT-1015 item
- rules of operation, control and maintenance of the National Instruments FPT-1015 item
- safety signs and mark certificates which confirm compatibility with appropriate standards
Why don't we read the manuals?
Usually it results from the lack of time and certainty about functionalities of purchased items. Unfortunately, networking and start-up of National Instruments FPT-1015 alone are not enough. An instruction contains a number of clues concerning respective functionalities, safety rules, maintenance methods (what means should be used), eventual defects of National Instruments FPT-1015, and methods of problem resolution. Eventually, when one still can't find the answer to his problems, he will be directed to the National Instruments service. Lately animated manuals and instructional videos are quite popular among customers. These kinds of user manuals are effective; they assure that a customer will familiarize himself with the whole material, and won't skip complicated, technical information of National Instruments FPT-1015.
Why one should read the manuals?
It is mostly in the manuals where we will find the details concerning construction and possibility of the National Instruments FPT-1015 item, and its use of respective accessory, as well as information concerning all the functions and facilities.
After a successful purchase of an item one should find a moment and get to know with every part of an instruction. Currently the manuals are carefully prearranged and translated, so they could be fully understood by its users. The manuals will serve as an informational aid.
Table of contents for the manual
FP T -1015 User Manual FPT -1015 User Manual December 2006 372051A-01[...]
Support Worldwide Technical Support and Product Info rmation ni.com National Instruments Corporate H[...]
Important Information Warranty The FPT-1015 is warranted against defects in materials and workmanshi[...]
Compliance Compliance with FCC/Canada Radio Frequency Interference Regulations Determining FCC Class[...]
Conventions The follo wing con ventions are used in this manual: » The » symbol leads you through [...]
© National Instruments Corporation vii FPT-1015 User Manual Contents Chapter 1 General Information [...]
© National Instruments Corporatio n 1-1 FPT -1015 User Manual 1 General Information This chapter in[...]
Chapter 1 General Information FPT -1015 User Manual 1-2 ni.com USB T ype B T ouchscreen Connector Th[...]
© National Instruments Corporatio n 2-1 FPT -1015 User Manual 2 Mounting This chapter explains how [...]
Chapter 2 Mountin g FPT -1015 User Manual 2-2 ni.com If one of the following situations ar ises, ha [...]
Chapter 2 Mounting © National Instruments Corporatio n 2-3 FPT -1015 User Manual W all Mounting The[...]
Chapter 2 Mountin g FPT -1015 User Manual 2-4 ni.com Panel Mounting Refer to Figure 2-2 and follow t[...]
Chapter 2 Mounting © National Instruments Corporatio n 2-5 FPT -1015 User Manual Rack Mounting The [...]
Chapter 2 Mountin g FPT -1015 User Manual 2-6 ni.com Desktop Stand Mounting The desktop stand kit is[...]
© National Instruments Corporation A-1 FPT-1015 User Manual A Specifications Physical Construction [...]
Appendix A Specifications FPT-1015 User Manu al A-2 ni.com Dimensions 383 [15.07] 307 [12.08] 372.6 [...]
Appendix A Specifications © National Instruments Corporation A-3 FPT-1015 User Manual T ouchscreen [...]
Appendix A Specifications FPT-1015 User Manu al A-4 ni.com Sensor board ......................... ..[...]
Appendix A Specifications © National Instruments Corporation A-5 FPT-1015 User Manual Environmental[...]
Appendix A Specifications FPT-1015 User Manu al A-6 ni.com Electromagnetic Compatibility This produc[...]
© National Instruments Corporation B-1 FPT-1015 User Manual B Display T iming Mode and OSD Supporte[...]
Appendix B Display T iming Mode an d OSD FPT-1015 User Manu al B-2 ni.com Keypad Interface T able B-[...]
Appendix B Display T iming Mode and OSD © National Instruments Corporation B-3 FPT-1015 User Manual[...]
© National Instruments Corporation C-1 FPT-1015 User Manual C T ouchscreen This appendix explains h[...]
Appendix C T ouchscreen FPT-1015 User Manu al C-2 ni.com Standard Calibration When you click this bu[...]
Appendix C T ouchscreen © National Instruments Corporation C-3 FPT-1015 User Manual Follo w these s[...]
Appendix C T ouchscreen FPT-1015 User Manu al C-4 ni.com Click on T ouch—This option all ows the m[...]
Appendix C T ouchscreen © National Instruments Corporation C-5 FPT-1015 User Manual Configuring the[...]
Appendix C T ouchscreen FPT-1015 User Manu al C-6 ni.com PenMount Monitor Menu Icon The PenMount Mon[...]
© National Instruments Corporatio n D-1 FPT -1015 User Manual D T echnical Support and Professional[...]
© National Instruments Corporation I-1 FPT-1015 User Manual Index A About function, C-5 adv anced c[...]
Index FPT-1015 User Manu al I-2 ni.com mounting, 2-1 desktop stand mounting , 2-6 panel mounting, 2 [...]
Index © National Instruments Corporation I-3 FPT-1015 User Manual PenMount Monitor menu icon, C-6 P[...]