Braun HNS 11 manual
- Read online or download the manual
- 28 pages
- 0.26 mb
Go to page of
Similar user manuals
Braun SERIES 3 390CC-312 pages 4.02 mb
Braun SI 62404 pages 2.17 mb
Braun ER 138332 pages 0.51 mb
Braun 11014 pages 4.64 mb
Braun G 300034 pages 3.48 mb
Blood Pressure Monitor
Braun BP200532 pages 0.59 mb
Braun Flex Integral 55255 pages 0.2 mb
Braun PV 22025 pages 0.15 mb
A good user manual
The rules should oblige the seller to give the purchaser an operating instrucion of Braun HNS 11, along with an item. The lack of an instruction or false information given to customer shall constitute grounds to apply for a complaint because of nonconformity of goods with the contract. In accordance with the law, a customer can receive an instruction in non-paper form; lately graphic and electronic forms of the manuals, as well as instructional videos have been majorly used. A necessary precondition for this is the unmistakable, legible character of an instruction.
What is an instruction?
The term originates from the Latin word „instructio”, which means organizing. Therefore, in an instruction of Braun HNS 11 one could find a process description. An instruction's purpose is to teach, to ease the start-up and an item's use or performance of certain activities. An instruction is a compilation of information about an item/a service, it is a clue.
Unfortunately, only a few customers devote their time to read an instruction of Braun HNS 11. A good user manual introduces us to a number of additional functionalities of the purchased item, and also helps us to avoid the formation of most of the defects.
What should a perfect user manual contain?
First and foremost, an user manual of Braun HNS 11 should contain:
- informations concerning technical data of Braun HNS 11
- name of the manufacturer and a year of construction of the Braun HNS 11 item
- rules of operation, control and maintenance of the Braun HNS 11 item
- safety signs and mark certificates which confirm compatibility with appropriate standards
Why don't we read the manuals?
Usually it results from the lack of time and certainty about functionalities of purchased items. Unfortunately, networking and start-up of Braun HNS 11 alone are not enough. An instruction contains a number of clues concerning respective functionalities, safety rules, maintenance methods (what means should be used), eventual defects of Braun HNS 11, and methods of problem resolution. Eventually, when one still can't find the answer to his problems, he will be directed to the Braun service. Lately animated manuals and instructional videos are quite popular among customers. These kinds of user manuals are effective; they assure that a customer will familiarize himself with the whole material, and won't skip complicated, technical information of Braun HNS 11.
Why one should read the manuals?
It is mostly in the manuals where we will find the details concerning construction and possibility of the Braun HNS 11 item, and its use of respective accessory, as well as information concerning all the functions and facilities.
After a successful purchase of an item one should find a moment and get to know with every part of an instruction. Currently the manuals are carefully prearranged and translated, so they could be fully understood by its users. The manuals will serve as an informational aid.
Table of contents for the manual
Stimuplex ® HNS 1 1 Nerve Stimulator for Regional Anaesthesia, Regional Analgesia, Neurology User m[...]
Distributor: B. Braun Melsungen AG P .O . Box 1 1 20 D-34209 Melsungen Tel. (0 56 6 1) 7 1-0 Fax (0 [...]
Contents P age Checks before start-up ..............................................................[...]
Checks before start-up The Stimuplex ® HNS 1 1 is a class lla medical device according to Council D[...]
In the operating mode, the LCD display shows the preselected stimulation current set on the controll[...]
Technical description 6. Refer to the Safety information section before using the Stimu- plex ® HNS[...]
Operating the nerve stimulator Switching on / switching off Turn the control knob clockwise out of t[...]
Selecting the current range The maximum current available can be selected by pressing the ”mA (max[...]
The LED displays All LEDs directly assigned to the buttons are used to monitor the selected operatin[...]
Pulse LED display The small orange LED above the LCD display flashes according to the pulse frequenc[...]
The tone Each generated pulse produces a tick . When a stimulation current flows to the electrodes ([...]
Special technical features The configuration of the electrode connection provides protection against[...]
Battery The battery must be regularly checked for correct charge by press- ing the ”BAT” button.[...]
Cleaning and disinfecting the Stimuplex ® HNS 1 1 Use only soft, damp cloths to clean and disinfect[...]
For usage of HNS 1 1 please refer to the individual local law restricts (eg. maintaining a logbook, [...]
Caution Electromagnetic fields may interfere with the function of the nerve stimulator . Do not use [...]
Safety information The nerve stimulator should not be used on patients with implanted electrical dev[...]
Instruments and accessories The Stimuplex ® HNS 1 1 nerve stimulator should only be operated with t[...]
Product discription Canula Code number Sales unit Ø x length Stimuplex ® A needles , 30 ° bevel A[...]
P eripheral electrical nerve stimulation Principles By simplifying the accurate location of peripher[...]
early detection of technical faults which could endanger the patient and place the success of the an[...]
If the two values coincide, the total resistance is acceptable (impedance test) and the electric cir[...]
Literature 23 Literature 1. Moore DC (1965), Regional block. A handbook for use in the clinical prac[...]
Appendix A Stimulation current display in the mA range Patient current Measurement display Controlle[...]
Appendix C 25 Stimuplex ® HNS 1 1 Nerve Stimulator Max. stimulation current as a function of tissue[...]
Stimuplex ® D and Contiplex ® D needles (DP: DE 39 19666) Only 0.2 or 0.05 mA stimulation current [...]
B. Braun Melsungen AG P .O. Box 1 1 20 D-34209 Melsungen Tel (0 56 6 1) 7 1- 0 Fax (0 56 61) 7 1- 38[...]