National Instruments FP-AI-112 manual
- Read online or download the manual
- 16 pages
- 0.39 mb
Go to page of
A good user manual
The rules should oblige the seller to give the purchaser an operating instrucion of National Instruments FP-AI-112, along with an item. The lack of an instruction or false information given to customer shall constitute grounds to apply for a complaint because of nonconformity of goods with the contract. In accordance with the law, a customer can receive an instruction in non-paper form; lately graphic and electronic forms of the manuals, as well as instructional videos have been majorly used. A necessary precondition for this is the unmistakable, legible character of an instruction.
What is an instruction?
The term originates from the Latin word „instructio”, which means organizing. Therefore, in an instruction of National Instruments FP-AI-112 one could find a process description. An instruction's purpose is to teach, to ease the start-up and an item's use or performance of certain activities. An instruction is a compilation of information about an item/a service, it is a clue.
Unfortunately, only a few customers devote their time to read an instruction of National Instruments FP-AI-112. A good user manual introduces us to a number of additional functionalities of the purchased item, and also helps us to avoid the formation of most of the defects.
What should a perfect user manual contain?
First and foremost, an user manual of National Instruments FP-AI-112 should contain:
- informations concerning technical data of National Instruments FP-AI-112
- name of the manufacturer and a year of construction of the National Instruments FP-AI-112 item
- rules of operation, control and maintenance of the National Instruments FP-AI-112 item
- safety signs and mark certificates which confirm compatibility with appropriate standards
Why don't we read the manuals?
Usually it results from the lack of time and certainty about functionalities of purchased items. Unfortunately, networking and start-up of National Instruments FP-AI-112 alone are not enough. An instruction contains a number of clues concerning respective functionalities, safety rules, maintenance methods (what means should be used), eventual defects of National Instruments FP-AI-112, and methods of problem resolution. Eventually, when one still can't find the answer to his problems, he will be directed to the National Instruments service. Lately animated manuals and instructional videos are quite popular among customers. These kinds of user manuals are effective; they assure that a customer will familiarize himself with the whole material, and won't skip complicated, technical information of National Instruments FP-AI-112.
Why one should read the manuals?
It is mostly in the manuals where we will find the details concerning construction and possibility of the National Instruments FP-AI-112 item, and its use of respective accessory, as well as information concerning all the functions and facilities.
After a successful purchase of an item one should find a moment and get to know with every part of an instruction. Currently the manuals are carefully prearranged and translated, so they could be fully understood by its users. The manuals will serve as an informational aid.
Table of contents for the manual
FieldPoint ™ Operating Instructions FP-AI-112 and cFP-AI-112 16-Channel, 16-Bit Anal og Input Modu les These operating instructions descri be ho w to install and use the FP-AI-112 and cFP-AI-112 analog input modules (referred to inclusiv ely as the [c]FP-A I-112). Fo r information about configuring and accessing the [c]F P- AI-112 ov er a network[...]
FP-AI-112 and cFP-AI-112 2 ni.com T o install the FP- AI-112, refer to Figure 1 and complete the follow ing steps: 1. Slide the terminal base key to either position X (used for an y module) or position 1 (used for the FP- AI-112). 2. Align the FP- AI-112 alignmen t sl ots with the guide rails on the terminal base. 3. Press firmly to seat the FP- AI[...]
© National Instruments Corp. 3 FP-AI-112 and cFP-AI-112 3. Using a number 2 Phillips scre wdriv er with a shank of at least 64 mm (2.5 in.) length, tighten the captiv e screws to 1. 1 N ⋅ m (10 lb ⋅ in.) of torque. The n ylon coating on the screws pre vents them from loosening. Figure 2. Installing the cFP-AI-112 Wiring the [c]FP-AI-112 The FP[...]
FP-AI-112 and cFP-AI-112 4 ni.com Y ou can use an external power supply to po wer field de vices. Connect the positi ve lead of the e xternal power supply to both V terminals and the negat ive lead to both C termina ls. Ensure that the current through both V terminals can nev er exceed 2 A. Install 2 A maximum, fast-acting fuses between the e xtern[...]
© National Instruments Corp. 5 FP-AI-112 and cFP-AI-112 Caution Cascading power between two modules defeats isolation between those modules. Cascading power from the network module defeats all isolation between modules in the FieldPoint bank. Measuring V oltage with the [c]FP-AI-112 Figure 3 sho ws how to connect a v oltage source without an exter[...]
FP-AI-112 and cFP-AI-112 6 ni.com Input Ranges The input ranges for voltage signals are 0 –1 V , 0 –5 V , 0 –10 V , ± 60 mV , ± 300 mV , ± 1V , ± 5 V , and ± 10 V . T o prevent inaccurate readings, choose an input range such that the signal you are measuring does not exceed either end of the rang e. Signals that greatly ex ceed the confi[...]
© National Instruments Corp. 7 FP-AI-112 and cFP-AI-112 ( number of input range s with 60 Hz filter ) × 0.10 s + ( number of channels with 500 Hz filter ) × 0.007 s + ( number of input ranges with 500 Hz filter ) × 0.012 s + 0.16 s = Update Period For the f astest performance, conf igure all unused channels for the 500 Hz f ilter setting and fo[...]
FP-AI-112 and cFP-AI-112 8 ni.com Isolation and Sa fety Guidelines Caution Read the following in formation before attempting to connect the [c]F P-AI-112 to an y circuits that may contain hazardous voltages. This section describes the isolat ion of the [c]FP-AI-112 and its compliance wit h international sa fety standards. The field wiring connectio[...]
© National Instruments Corp. 9 FP-AI-112 and cFP-AI-112 using 14 A WG (1.6 mm) wire with a ring lug. Use the 5/16 in. panhead screw shipped with the backplane to secure the ring lug to the backplane PE ground terminal. • As with any hazardous v oltage wiring, make sure that all wiring and connections meet appl icable electrical codes and commons[...]
FP-AI-112 and cFP-AI-112 10 ni.com Effecti ve resolution by inpu t range and filter setting Update period and input band width by filter setting Normal-mode rejection ............. ........ 95 dB Nonlinearity .......... ........... ......... ........ 0.0015% (monotonicity 1 warranted ov er the operating temperature range) Input impedance (with powe[...]
© National Instruments Corp. 11 FP-AI-112 and cFP-AI-112 T ypical and warranted accuracy b y input range and temperature range Note Full scale is the maximum value of the nominal input range. F or example, for the ± 10 V input range, full scale is 10 V and ± 0.01% of full scale is 1 mV . Gain error drift ...... ......... ........... ........ ± [...]
FP-AI-112 and cFP-AI-112 12 ni.com Physical Char acteristics Indicators ................... ......... ........... ... Green POWER and READY indicators W eight FP-AI-112..... ............. ............ ..... 142 g (5.0 oz) cFP-AI-112 ....... ............ .............. 113 g (4.0 oz) Power Requirements Power from network module ............ 350 mW I[...]
© National Instruments Corp. 13 FP-AI-112 and cFP-AI-112 Shock and Vibration These specifications apply only to the cFP-AI-112. NI recommends Compact FieldP oint if your application is subject to shock and vibration. Operating vibration, random (IEC 60068-2-64)....... ........... ............ 1 0 –500 Hz, 5 g rms Operating vibration, sinusoidal [...]
FP-AI-112 and cFP-AI-112 14 ni.com CE Compliance This product meets the essentia l requirements of applicable European Directi ves, as amended for CE ma rking, as follows: Lo w-V oltage Directi ve (s afety)...... ... 73/23/EEC Electromagnet ic Compatibility Directiv e (EMC) ... ........... ......... ........ 89/336/EEC Note Refer to the Declaration[...]
© National Instruments Corp. 15 FP-AI-112 and cFP-AI-112 Where to Go for Support For more information about settin g up the FieldPoint system, refer to these National Instruments documents: • FieldPoint network module user manual • Other FieldPo int I/O m odule operating instructions • FieldPoint terminal base and connector block operating i[...]
© 2004 National I nstruments Corp. All rights reser ved. 371330A-01 Jul04 *371330A-01* FieldPoint™, Nati onal Instruments™, NI™, and ni.com™ are trademarks of National Instruments Corporati on. Product and company names ment ioned herein are trademarks or trade names of thei r respective companies. For patents covering National Instruments[...]