National Instruments EIGHT CHANNEL SPST RELAY MODULE manual
- Read online or download the manual
- 14 pages
- 0.35 mb
Go to page of
Similar user manuals
National Instruments EIGHT CHANNEL SPST RELAY MODULE14 pages 0.35 mb
National Instruments NI-845x Software and Hardware6 pages 0.09 mb
National Instruments NI-845x6 pages 0.09 mb
National Instruments CFP-RLY-42114 pages 0.35 mb
A good user manual
The rules should oblige the seller to give the purchaser an operating instrucion of National Instruments EIGHT CHANNEL SPST RELAY MODULE, along with an item. The lack of an instruction or false information given to customer shall constitute grounds to apply for a complaint because of nonconformity of goods with the contract. In accordance with the law, a customer can receive an instruction in non-paper form; lately graphic and electronic forms of the manuals, as well as instructional videos have been majorly used. A necessary precondition for this is the unmistakable, legible character of an instruction.
What is an instruction?
The term originates from the Latin word „instructio”, which means organizing. Therefore, in an instruction of National Instruments EIGHT CHANNEL SPST RELAY MODULE one could find a process description. An instruction's purpose is to teach, to ease the start-up and an item's use or performance of certain activities. An instruction is a compilation of information about an item/a service, it is a clue.
Unfortunately, only a few customers devote their time to read an instruction of National Instruments EIGHT CHANNEL SPST RELAY MODULE. A good user manual introduces us to a number of additional functionalities of the purchased item, and also helps us to avoid the formation of most of the defects.
What should a perfect user manual contain?
First and foremost, an user manual of National Instruments EIGHT CHANNEL SPST RELAY MODULE should contain:
- informations concerning technical data of National Instruments EIGHT CHANNEL SPST RELAY MODULE
- name of the manufacturer and a year of construction of the National Instruments EIGHT CHANNEL SPST RELAY MODULE item
- rules of operation, control and maintenance of the National Instruments EIGHT CHANNEL SPST RELAY MODULE item
- safety signs and mark certificates which confirm compatibility with appropriate standards
Why don't we read the manuals?
Usually it results from the lack of time and certainty about functionalities of purchased items. Unfortunately, networking and start-up of National Instruments EIGHT CHANNEL SPST RELAY MODULE alone are not enough. An instruction contains a number of clues concerning respective functionalities, safety rules, maintenance methods (what means should be used), eventual defects of National Instruments EIGHT CHANNEL SPST RELAY MODULE, and methods of problem resolution. Eventually, when one still can't find the answer to his problems, he will be directed to the National Instruments service. Lately animated manuals and instructional videos are quite popular among customers. These kinds of user manuals are effective; they assure that a customer will familiarize himself with the whole material, and won't skip complicated, technical information of National Instruments EIGHT CHANNEL SPST RELAY MODULE.
Why one should read the manuals?
It is mostly in the manuals where we will find the details concerning construction and possibility of the National Instruments EIGHT CHANNEL SPST RELAY MODULE item, and its use of respective accessory, as well as information concerning all the functions and facilities.
After a successful purchase of an item one should find a moment and get to know with every part of an instruction. Currently the manuals are carefully prearranged and translated, so they could be fully understood by its users. The manuals will serve as an informational aid.
Table of contents for the manual
323617A-01 May 2003 © 2003 National I nstruments Corp. Al l rights res erved. FieldPoint™, Nati onal Instruments™, NI™, and ni.com™ are trademarks of National Instruments Corporation. Product and company names ment ioned herein are trademark s or trade names of their respecti ve companies. For patents covering National Inst ruments product[...]
cFP-RLY-421 2 ni.com Before you conf igure a FieldPoint system that uses a cFP-RL Y -421 module, calculate the total power consumption of the I/O modules on the FieldPoint bank. Y ou ca n find the po wer requirement specification in the operating inst ructions for ea ch I/O module . The maximum power the netw ork module can supply is specified in t[...]
© National Instruments Corp. 3 cFP-RLY-421 Figure 1. Installing the cFP-RL Y -421 Wiring the cFP-RL Y -421 The cFP-CB- x connector block has conne ctions for each of the eight cFP-RL Y -421 relay channels and for an external supply to power f ield devices. If you are using the cFP-RL Y - 421 in a hazardous voltage application, you mu st use the cF[...]
cFP-RLY-421 4 ni.com T able 1 li sts the terminal assign ments for the signals of each channel. All of the COM terminals are conne cted internally and all of the V SUP terminals are connected inte rnally . NI does not recommend using them with the cFP-RL Y -421. Caution Cascading power between two modules defeats isolation between those modules. Ca[...]
© National Instruments Corp. 5 cFP-RLY-421 Connecting Loads to the cFP-RL Y -421 W ire an external po wer supply to the load and IC terminal of the individu al channel as shown in Figure 2. This method of wiring keeps the electrical connections of the cFP-RL Y -421 within safe current limits. Figure 2. Connecting a Load The cFP-RL Y -421 has eight[...]
cFP-RLY-421 6 ni.com In addition, the cFP-RL Y -421 has internal protection MO Vs to pre vent excessiv ely high voltag e from being applied across the contacts. The MO Vs are located between the NO and IC contacts of each relay . Howe ver , National Instruments still recommends the use of a protection circuit across an inductive load. The flyback p[...]
© National Instruments Corp. 7 cFP-RLY-421 Diode Circuit CR Circuits Varistor Circuit Diagram Notes Use in DC applications only . Compared to RC type, circuit delays release time (2 to 5 times values stated in catalog). For lar ger voltages, use diode with rev erse breakdo wn 10 times circu it voltage an d forward load circuit. For smaller v oltag[...]
cFP-RLY-421 8 ni.com In-Rush Current The type of l oad and its in-rush current charac teristics, together with switching frequency , can ca use contact welding. For loads with in-rush current, measure the steady state current and in-rush current to determine the proper rela y . Some typical types of loads and the in-rush current they create are sum[...]
© National Instruments Corp. 9 cFP-RLY-421 Upgrading the FieldPoint Firmware Y ou may need to upgrade the FieldPoint f irmware when you add ne w I/O modules to the FieldPoi nt system. For information on determining which fi rmware you need and ho w to upgrade the firmw are, go to ni.com/info and enter fpmatrix . Isolation and Sa fety Guidelines Ca[...]
cFP-RLY-421 10 ni.com •D o not share the exter nal supply voltages (the V and C terminals) with other de vices (including othe r FieldPoint de vices), unless those devices ar e isolated from human contact. • Y ou must connect the protecti ve ea rth (PE) ground terminal on the cFP-BP- x backplane to the syst em safety ground. The backplane PE gr[...]
© National Instruments Corp. 11 cFP-RLY-421 Specifications The follo wing specifications ar e typical for a range of – 40 to 60 °C unless otherwise noted. Relay Characteristics Number of channels ..................... ..... 8 Relay type ......... ...................... .......... 1 SPST , nonlatching Maximum switching ca pacity (resisti ve load[...]
cFP-RLY-421 12 ni.com Physical Indicators ..... .............. ........... ........... . Green POWER and READY indicators, 8 green output state indicators W eight cFP-RL Y -421 .................... .......... 130 g (4.5 oz) Power Requirements Power from network modul e ............ 1,700 mW Isolation Voltage Maximum isolation v oltage .............[...]
© National Instruments Corp. 13 cFP-RLY-421 Safety This product is designed to meet the requirements of the following standards of safety for electric al equipment for measurement, control, and laboratory use: • IEC 61010-1, EN 61010-1 • UL 3121-1, UL 61010C-1 • CAN/CSA C22.2 No. 1010.1 For UL, hazardous location, and othe r safety certifica[...]
cFP-RLY-421 14 ni.com Where to Go for Support For more information about settin g up the FieldPoint system, refer to these National Instruments documents: • FieldPoint network module user manual • Other FieldPo int I/O m odule operating instructions • FieldPoint connector bl ock operating instructions Go to ni.com/support for the most current[...]