National Instruments cFP-Al-110 manual
- Read online or download the manual
- 20 pages
- 0.41 mb
Go to page of
A good user manual
The rules should oblige the seller to give the purchaser an operating instrucion of National Instruments cFP-Al-110, along with an item. The lack of an instruction or false information given to customer shall constitute grounds to apply for a complaint because of nonconformity of goods with the contract. In accordance with the law, a customer can receive an instruction in non-paper form; lately graphic and electronic forms of the manuals, as well as instructional videos have been majorly used. A necessary precondition for this is the unmistakable, legible character of an instruction.
What is an instruction?
The term originates from the Latin word „instructio”, which means organizing. Therefore, in an instruction of National Instruments cFP-Al-110 one could find a process description. An instruction's purpose is to teach, to ease the start-up and an item's use or performance of certain activities. An instruction is a compilation of information about an item/a service, it is a clue.
Unfortunately, only a few customers devote their time to read an instruction of National Instruments cFP-Al-110. A good user manual introduces us to a number of additional functionalities of the purchased item, and also helps us to avoid the formation of most of the defects.
What should a perfect user manual contain?
First and foremost, an user manual of National Instruments cFP-Al-110 should contain:
- informations concerning technical data of National Instruments cFP-Al-110
- name of the manufacturer and a year of construction of the National Instruments cFP-Al-110 item
- rules of operation, control and maintenance of the National Instruments cFP-Al-110 item
- safety signs and mark certificates which confirm compatibility with appropriate standards
Why don't we read the manuals?
Usually it results from the lack of time and certainty about functionalities of purchased items. Unfortunately, networking and start-up of National Instruments cFP-Al-110 alone are not enough. An instruction contains a number of clues concerning respective functionalities, safety rules, maintenance methods (what means should be used), eventual defects of National Instruments cFP-Al-110, and methods of problem resolution. Eventually, when one still can't find the answer to his problems, he will be directed to the National Instruments service. Lately animated manuals and instructional videos are quite popular among customers. These kinds of user manuals are effective; they assure that a customer will familiarize himself with the whole material, and won't skip complicated, technical information of National Instruments cFP-Al-110.
Why one should read the manuals?
It is mostly in the manuals where we will find the details concerning construction and possibility of the National Instruments cFP-Al-110 item, and its use of respective accessory, as well as information concerning all the functions and facilities.
After a successful purchase of an item one should find a moment and get to know with every part of an instruction. Currently the manuals are carefully prearranged and translated, so they could be fully understood by its users. The manuals will serve as an informational aid.
Table of contents for the manual
FieldPoint ™ Operating Instructions FP-AI-110 and cFP-AI-110 Eight-Channel, 16-Bit Analog Input M odules These operating instructions descri be ho w to install and use the FP-AI-110 and cFP-AI-110 analog input modules (referred to inclusiv ely as the [c]FP-A I-110). F or information about configuring and accessing the [c]F P- AI-110 ov er a netwo[...]
FP-AI-110 and cFP-AI-110 2 ni.com T o install the FP- AI-110, refer to Figure 1 and complete the follow ing steps: 1. Slide the terminal base key to either position X (used for an y module) or position 1 (used for the FP- AI-110). 2. Align the FP- AI-110 alignmen t sl ots with the guide rails on the terminal base. 3. Press firmly to seat the FP- AI[...]
© National Instruments Corp. 3 FP-AI-110 and cFP-AI-110 (10 lb ⋅ in.) of torque. The nylon coating on the screws pre vents them from loosening. Figure 2. Installing the cFP-AI-110 Wiring the [c]FP-AI-110 The FP-TB- x terminal base ha s connections for each of the eight input channels and for an external power su pply to power field de vice s. Th[...]
FP-AI-110 and cFP-AI-110 4 ni.com is 2 A or less and the maximum current through any V SUP terminal is 1 A or less . Install a 2 A maximum, fast-acting fuse between t he external power supply and the V terminal on each channel. The wiring diagrams in this document sho w fuses where appropriate. T able 1 lists the terminal assign ments for the signa[...]
© National Instruments Corp. 5 FP-AI-110 and cFP-AI-110 T aking Measurements with the [c]FP-AI-110 The [c]FP- AI-110 has eight single- ended input cha nnels. All eight channels share a common ground referen ce that is isolated from other modules in the FieldPoint system. Figure 3 shows the analog input circuitry on one channel. Figure 3. FP- AI-11[...]
FP-AI-110 and cFP-AI-110 6 ni.com Figure 5 sho ws how to connect a v oltage source with an extern al power supply to one channel of the [c]FP-AI-110. Figure 5. Voltage Source with Ex ternal Powe r Supply Measuring Current with the [c]FP-AI-110 The input ranges for current sources are 0 –20, 4 –20, and ± 20 mA. The module reads current flowing [...]
© National Instruments Corp. 7 FP-AI-110 and cFP-AI-110 Figure 6. Current Source without External Power Supply Figure 7 sho ws how to connect a cu rrent sour ce with an external power supply to one channel of the [c]FP-AI-110. Figure 7. Current Source with Ex ternal Power Supply Input Ranges T o prev ent inaccurate readings, c hoose an input range[...]
FP-AI-110 and cFP-AI-110 8 ni.com Overranging The [c]FP- AI-110 has an overranging feature that meas ures a little beyond the nominal values of each rang e. For example, the actual measurement limit of the ± 10 V range is ± 10.4 V . The overrang ing feature enables the [c]FP- AI-110 to compensate for field de vices with span errors of up to + 4% [...]
© National Instruments Corp. 9 FP-AI-110 and cFP-AI-110 sampling rate is faster than th e rate at which the network module polls the [c]FP-AI-110 for data. Status Indicators The [c]FP-AI-110 has tw o green status LEDs, POWER and READY . After you insert the [c] FP-AI-110 into a terminal base or backplane and apply po wer to th e connected network [...]
FP-AI-110 and cFP-AI-110 10 ni.com working v oltages of 250 V rms 1 . Safety standards (such as those published by UL and IEC) requir e the use of double insulation between hazardous voltages and an y human-accessi ble parts or circuits. Never try to use any isolation produ ct between human-accessible parts (such as DIN rails or monito ring station[...]
© National Instruments Corp. 11 FP-AI-110 and cFP-AI-110 • Operate the [c]FP-AI-110 only at or belo w Pollution Degree 2. Pollution Degree 2 means th at only nonconductiv e pollution occurs in most case s. Occasionally , howe ver , a temporary conductivity caused by conde nsation must be e xpected. • Operate the [c]FP-AI-110 at or below Measur[...]
FP-AI-110 and cFP-AI-110 12 ni.com Safety Guidelines for Hazardous V oltages If hazardous voltages are connected to the module, take the follow ing precautions. A hazardous voltage is a v oltage greater than 42.4 V peak or 60 VDC to earth ground. Caution Ensure that hazardous voltage wiring is performed only by qualified personnel adhering to local[...]
© National Instruments Corp. 13 FP-AI-110 and cFP-AI-110 Effecti ve resolution by inpu t signal range and f ilter setting Input charac teristics by f ilter setting Normal-mode rejection .. .............. ..... 95 dB (with 50/60 Hz filter) Nonlinearity .......... ........... ......... ........ 0.0015% (monotonicity 1 guaranteed ov er the operating [...]
FP-AI-110 and cFP-AI-110 14 ni.com Input current 25 °C........... ...................... .......... 400 pA typ, 1 nA max 70 °C ............. ........... ........... ........ 3 nA typ, 15 nA max Input noise (with 50 or 60 Hz filter enabled) ± 60 mV range ...... ........... ........... . ± 3 LSB 1 peak-to-peak ± 300 mV range ...... ........... .[...]
© National Instruments Corp. 15 FP-AI-110 and cFP-AI-110 Note Full scale is the maximum value of the nominal input range. F or example, for the ± 10 V input range, full scale is 10 V and ± 0.01% of full scale is 1 mV . Gain error drift ........... ......... ........... ... ± 20 ppm/°C Offset error drift W ith 50 or 60 Hz filte r enabled ......[...]
FP-AI-110 and cFP-AI-110 16 ni.com Physical Char acteristics Indicators ............ ........... ........... ........ Green POWER and READY indicators W eight FP-AI-110..... ............. .............. ... 140 g (4.8 oz) cFP-AI-110 ....... ............ .............. 110 g (3.7 oz) Power Requirements Power from network module ............ 350 mW S[...]
© National Instruments Corp. 17 FP-AI-110 and cFP-AI-110 Operating shock (IEC 60068-2-27)........... ........... ........ 50 g, 3 ms half sine, 18 shocks at 6 orientations; 30 g, 11 ms half sine, 18 shocks at 6 orientations Safety This product is designed to meet the requirements of the following standards of safety for electric al equipment for m[...]
FP-AI-110 and cFP-AI-110 18 ni.com CE Compliance This product meets the essentia l requirements of applicable European Directi ves, as amended for CE ma rking, as follows: Lo w-V oltage Directi ve (s afety). ..... ... 73/23/EEC Electromagnet ic Compatibility Directiv e (EMC) ... ........... ......... ........ 89/336/EEC Note Refer to the Declaratio[...]
© National Instruments Corp. 19 FP-AI-110 and cFP-AI-110 Where to Go for Support For more information about settin g up the FieldPoint system, refer to these National Instruments documents: • FieldPoint network module user manual • Other FieldPo int I/O m odule operating instructions • FieldPoint terminal base and connector block operating i[...]