National Instruments SHC68-H1X38 manual
- Read online or download the manual
- 8 pages
- 0.4 mb
Go to page of
Similar user manuals
National Instruments NI USB-625912 pages 0.48 mb
National Instruments NI 923436 pages 0.55 mb
National Instruments Order Analysis Toolset63 pages 0.75 mb
National Instruments LM1085 Series13 pages 0.73 mb
National Instruments NI 942522 pages 0.3 mb
National Instruments NI PCIe-836271 pages 1.49 mb
National Instruments NI PXI-810883 pages 3.27 mb
National Instruments MCA-77249 pages 0.08 mb
A good user manual
The rules should oblige the seller to give the purchaser an operating instrucion of National Instruments SHC68-H1X38, along with an item. The lack of an instruction or false information given to customer shall constitute grounds to apply for a complaint because of nonconformity of goods with the contract. In accordance with the law, a customer can receive an instruction in non-paper form; lately graphic and electronic forms of the manuals, as well as instructional videos have been majorly used. A necessary precondition for this is the unmistakable, legible character of an instruction.
What is an instruction?
The term originates from the Latin word „instructio”, which means organizing. Therefore, in an instruction of National Instruments SHC68-H1X38 one could find a process description. An instruction's purpose is to teach, to ease the start-up and an item's use or performance of certain activities. An instruction is a compilation of information about an item/a service, it is a clue.
Unfortunately, only a few customers devote their time to read an instruction of National Instruments SHC68-H1X38. A good user manual introduces us to a number of additional functionalities of the purchased item, and also helps us to avoid the formation of most of the defects.
What should a perfect user manual contain?
First and foremost, an user manual of National Instruments SHC68-H1X38 should contain:
- informations concerning technical data of National Instruments SHC68-H1X38
- name of the manufacturer and a year of construction of the National Instruments SHC68-H1X38 item
- rules of operation, control and maintenance of the National Instruments SHC68-H1X38 item
- safety signs and mark certificates which confirm compatibility with appropriate standards
Why don't we read the manuals?
Usually it results from the lack of time and certainty about functionalities of purchased items. Unfortunately, networking and start-up of National Instruments SHC68-H1X38 alone are not enough. An instruction contains a number of clues concerning respective functionalities, safety rules, maintenance methods (what means should be used), eventual defects of National Instruments SHC68-H1X38, and methods of problem resolution. Eventually, when one still can't find the answer to his problems, he will be directed to the National Instruments service. Lately animated manuals and instructional videos are quite popular among customers. These kinds of user manuals are effective; they assure that a customer will familiarize himself with the whole material, and won't skip complicated, technical information of National Instruments SHC68-H1X38.
Why one should read the manuals?
It is mostly in the manuals where we will find the details concerning construction and possibility of the National Instruments SHC68-H1X38 item, and its use of respective accessory, as well as information concerning all the functions and facilities.
After a successful purchase of an item one should find a moment and get to know with every part of an instruction. Currently the manuals are carefully prearranged and translated, so they could be fully understood by its users. The manuals will serve as an informational aid.
Table of contents for the manual
USER GUIDE Digital Flying Lead Cable Accessor y for Single-Ended NI Digi tal W aveform Generator/Analyzers The NI SHC68-H1X38 cable is a singl e-ended cable that breaks out each NI digital wav eform generator/analyzer signal in to two 0.1 inch header receptacles, one receptacle each for the signal and ground. The NI SHC68-H1X38 cable provides an ea[...]
Digital Flying Lead Cable User Guide 2 ni.com What Y ou Need to Get Started To set up and use the NI SHC68-H1X38 cable, you need the following items: ❑ Compatible NI digi tal wav eform gene rator/analyzer , installed in a PXI chassis, compactPCI ch assis, or PCI slot ❑ The documentation included wi th the digital wav eform generator/analyzer an[...]
© National Instruments Corporation 3 Di gital Flying Lead Cable User Guide Parts Locator Refer to Figure 1 to locate connectors and comp onents on the NI SHC68-H1X38 cable. The lead pair s are shown in more d etail in Figure 2. Figure 1. NI SHC68-H1X38 Cable Parts Locator Diagram 1 Lead Pairs 2 Removable Sleeving 3 68-Pin VHDCI Connector 1 2 3[...]
Digital Flying Lead Cable User Guide 4 ni.com Figure 2. Detailed Parts Loc ator Diagram for the Lead Pairs Connecting Signals Each DIO, PFI, and clock channel of the NI digital waveform generator/analyzer connects to a co rresponding pair of leads on the NI SHC68-H1X38 cable. The NI SHC68-H1X38 cable is separated into two bundles of 19 channels, an[...]
© National Instruments Corporation 5 Di gital Flying Lead Cable User Guide CLK OUT / PFI 4 For de vices that have a DDC CLK OUT channel, this signal is the exported sample clock signal. Otherwise, th is is PFI 4. Bundle A PFI <1..3> Programmable functional interface PFI <1..3>. PFI 1: Bundle A PFI <2..3>: Bundle B Reserved / PFI [...]
Digital Flying Lead Cable User Guide 6 ni.com Using Y our Accessor y The NI SCH68-H1X38 cable breaks out every signal through a 50 Ω characteristic impedance microcoaxial cable, which is then split into the two leads with receptacles. This cab le was designed for both acquisition and generation operations with your single-ended NI digital wavefor[...]
© National Instruments Corporation 7 Di gital Flying Lead Cable User Guide Specifications Digital I/O DIO Channels .................... ..................... 32, single-ended Control I/O Channels ........... .................. 6, single-ended Typical Propagation delay ...... ............... 13.5 ns Typical channel-to-channel skew........... ±250 [...]
National Instruments, NI, ni.com, and LabVIEW are tradem arks of National Instruments Corporation. Refer to t he Terms of U se section on ni.com/legal for more in formation abou t National Instruments trademarks. Other product and compan y names mentioned herein are trademarks or trade names of their respective companie s. For patents covering Nati[...]