National Instruments SCXI -1125 manual
- Read online or download the manual
- 14 pages
- 0.14 mb
Go to page of
Similar user manuals
National Instruments NI 784xR74 pages 1.43 mb
National Instruments SCC-TC0116 pages 0.28 mb
National Instruments NI 9239E26 pages 0.42 mb
National Instruments PCI-485/819 pages 0.13 mb
National Instruments NI PXI-836038 pages 0.61 mb
National Instruments USB device 622x23 pages 0.85 mb
Frozen Dessert Maker
National Instruments 320727B-0150 pages 0.25 mb
National Instruments FP-QUAD-51012 pages 0.16 mb
A good user manual
The rules should oblige the seller to give the purchaser an operating instrucion of National Instruments SCXI -1125, along with an item. The lack of an instruction or false information given to customer shall constitute grounds to apply for a complaint because of nonconformity of goods with the contract. In accordance with the law, a customer can receive an instruction in non-paper form; lately graphic and electronic forms of the manuals, as well as instructional videos have been majorly used. A necessary precondition for this is the unmistakable, legible character of an instruction.
What is an instruction?
The term originates from the Latin word „instructio”, which means organizing. Therefore, in an instruction of National Instruments SCXI -1125 one could find a process description. An instruction's purpose is to teach, to ease the start-up and an item's use or performance of certain activities. An instruction is a compilation of information about an item/a service, it is a clue.
Unfortunately, only a few customers devote their time to read an instruction of National Instruments SCXI -1125. A good user manual introduces us to a number of additional functionalities of the purchased item, and also helps us to avoid the formation of most of the defects.
What should a perfect user manual contain?
First and foremost, an user manual of National Instruments SCXI -1125 should contain:
- informations concerning technical data of National Instruments SCXI -1125
- name of the manufacturer and a year of construction of the National Instruments SCXI -1125 item
- rules of operation, control and maintenance of the National Instruments SCXI -1125 item
- safety signs and mark certificates which confirm compatibility with appropriate standards
Why don't we read the manuals?
Usually it results from the lack of time and certainty about functionalities of purchased items. Unfortunately, networking and start-up of National Instruments SCXI -1125 alone are not enough. An instruction contains a number of clues concerning respective functionalities, safety rules, maintenance methods (what means should be used), eventual defects of National Instruments SCXI -1125, and methods of problem resolution. Eventually, when one still can't find the answer to his problems, he will be directed to the National Instruments service. Lately animated manuals and instructional videos are quite popular among customers. These kinds of user manuals are effective; they assure that a customer will familiarize himself with the whole material, and won't skip complicated, technical information of National Instruments SCXI -1125.
Why one should read the manuals?
It is mostly in the manuals where we will find the details concerning construction and possibility of the National Instruments SCXI -1125 item, and its use of respective accessory, as well as information concerning all the functions and facilities.
After a successful purchase of an item one should find a moment and get to know with every part of an instruction. Currently the manuals are carefully prearranged and translated, so they could be fully understood by its users. The manuals will serve as an informational aid.
Table of contents for the manual
CVI ™ , LabVIEW ™ , Natio nal Instru ments ™ ,n i . c o m ™ ,N I - D A Q ™ , and SCXI ™ are trademarks of National Instruments Corporation. Product a nd company names mentioned herein are tr ademarks or trade names of their respective companies. 370271A-01 © Copyright 2000 National Instruments Corp. All rights reserved. December 2000 C[...]
SCXI-1125 Calibration Procedure 2 ni.com Equipment and Other T est Requirements This section describes the equipment, software, documentation, and test conditions required for calibrat ing the SCXI-1125. T est Equipment Calibration requires a high-precisi on voltage source with at least 50 ppm accuracy, a multiranging 5 1/2 digit digital multimeter[...]
© National Instruments Corpor ation 3 SCXI-1125 Calibration Procedur e Documentation The NI-DAQ Function Refe rence Online Help fi le and NI-DAQ User Manual for PC Compa tibles c o n t a i nd e t a i l e di n f o r m a t i o no nu s i n gt h e NI-DAQ driver. The funct ion reference online h elp file includes detaile d information on the driver fun[...]
SCXI-1125 Calibration Procedure 4 ni.com Setting Up Y our Module Complete the following steps to set up yo ur module for testing : 1. Install the SCXI-1125 in sl ot 1 of the SCXI chassis. 2. Connect a 68-to-68-pin cable between the S CXI module and the E Series D A Q de vice installed in the host computer . 3. Configure the hardware with Measuremen[...]
© National Instruments Corpor ation 5 SCXI-1125 Calibration Procedur e 7. Connect the calibrator to the appro priate analog input, starting with channel 0. Refer to Figure 1 to determine which pins on the 96-pin front conn ector correspond to t he positiv e and negati ve inputs of the specified channel. Fo r example, the positive input for channel[...]
SCXI-1125 Calibration Procedure 6 ni.com 13. Call SCXI_Scale . Set the following parameters: • SCXIChassisID — The v alue assigned b y MAX • moduleSlot — 1 • channel — The same channel number from step 7 • SCXIgain — The same gain value from step 9 • TBgain — 1 • DAQboard — The device number for the D A Q device • DAQChann[...]
© National Instruments Corpor ation 7 SCXI-1125 Calibration Procedur e • calOp — ND_SELF_CALIBRATE • s etOfCalConstants — ND_USER_EEPROM_AREA • calRefVolts — 0.0 4. Call AI_Configure to put the E Series device in bipolar mode. Set t he following parameters: • deviceNumb er — The v alue assigned b y MAX • chan — 0 • inputMode [...]
SCXI-1125 Calibration Procedure 8 ni.com • filterMode — 1 • cutoffFrequency — 10,000 9. Connect the calibrator to the analog input you w ant to adjust. Refer to Figure 1 to determine the pins on the 96-pin f ront connector that correspond to the positiv e and neg ati ve inputs of the specif ied channel. For e xamp le, the positi ve input fo[...]
© National Instruments Corpor ation 9 SCXI-1125 Calibration Procedur e 16. Record binary1 with volt1 and binary2 with volt2 for later use. 17. Repeat steps 7 through 16 for the remaining gain v alues you want to adjust on this channel. 18. Repeat steps 6 through 17 for the remaining chan nels you want to adjust. Y ou hav e determined the new calib[...]
SCXI-1125 Calibration Procedure 10 ni.com V erifying Adjusted V alues After completing the adjustment s to your SCXI-1125 module, it is important that you verify the analog input operation by repeating t he steps listed in the Veri fying Gain and Offset Errors section of this docu ment. Reverifying after adjustments ensures that your SCXI-1125 mod [...]
© National Instruments Corpor ation 11 SCXI-1125 Calibration Procedur e Front and Rear Signal Connector Diagrams Figure 1. Front Signal Connector Pin Assignments CHSGND +5 V RSVD CH7+ CH6+ CH5+ CH4+ CH3+ CH2+ CH1+ NC NC NC NC RSVD CH0+ 2 4 6 8 10 12 14 16 18 20 22 24 26 28 30 32 2 4 6 8 10 12 14 16 18 20 22 24 26 28 30 32 AB RSVD CJTEMP RSVD RSVD [...]
SCXI-1125 Calibration Procedure 12 ni.com Figure 2. Rear Signal Connector Pin Assignments SERCLK DIGGND SLOT0SEL* DAQD*/A SERDATIN OUTREF MCH7+ MCH6+ MCH5+ MCH4+ MCH3+ MCH2+ MCH1+ MCH0+ AOGND SCANCLK SYNC SERDATOUT DIGGND MCH7 – MCH6 – MCH5 – MCH4 – MCH3 – MCH2 – MCH1 – MCH0 – AOGND 49 50 47 48 45 46 43 44 41 42 39 40 37 38 35 36 33[...]
© National Instruments Corpor ation 13 SCXI-1125 Calibration Procedur e Specifications The following table contains the sp ecifications you should use w hen verifying and calibrating y our SCXI-1125 module. If you r device has been calibrated within the last year, the test point value should fall between the Upper Limit and Lower Limit valu es. Ta[...]
SCXI-1125 Calibration Procedure 14 ni.com 200 – 0.025 – 0.024750 – 0.02468642 – 0 .02481358 250 0.02 0.019800 0.01985126 0.019 74874 250 0 0.000000 0.000022 – 0.000022 250 – 0.02 – 0.019800 – 0.01974874 – 0.0 1985126 500 0.01 0.009900 0.00992663 0.009 87337 500 0 0.000000 0.000012 – 0.000012 500 – 0.01 – 0.009900 – 0.00987[...]