National Instruments SCXI -1125 manual
- Read online or download the manual
- 14 pages
- 0.14 mb
Go to page of
Similar user manuals
National Instruments NI 1450 Series83 pages 1.49 mb
National Instruments DAQ 652747 pages 0.29 mb
National Instruments NI 510287 pages 0.67 mb
National Instruments NI 541226 pages 0.25 mb
National Instruments NI 926526 pages 0.41 mb
National Instruments FP-QUAD-51012 pages 0.16 mb
National Instruments 62117 pages 0.35 mb
National Instruments PCI-6025E132 pages 3.37 mb
A good user manual
The rules should oblige the seller to give the purchaser an operating instrucion of National Instruments SCXI -1125, along with an item. The lack of an instruction or false information given to customer shall constitute grounds to apply for a complaint because of nonconformity of goods with the contract. In accordance with the law, a customer can receive an instruction in non-paper form; lately graphic and electronic forms of the manuals, as well as instructional videos have been majorly used. A necessary precondition for this is the unmistakable, legible character of an instruction.
What is an instruction?
The term originates from the Latin word „instructio”, which means organizing. Therefore, in an instruction of National Instruments SCXI -1125 one could find a process description. An instruction's purpose is to teach, to ease the start-up and an item's use or performance of certain activities. An instruction is a compilation of information about an item/a service, it is a clue.
Unfortunately, only a few customers devote their time to read an instruction of National Instruments SCXI -1125. A good user manual introduces us to a number of additional functionalities of the purchased item, and also helps us to avoid the formation of most of the defects.
What should a perfect user manual contain?
First and foremost, an user manual of National Instruments SCXI -1125 should contain:
- informations concerning technical data of National Instruments SCXI -1125
- name of the manufacturer and a year of construction of the National Instruments SCXI -1125 item
- rules of operation, control and maintenance of the National Instruments SCXI -1125 item
- safety signs and mark certificates which confirm compatibility with appropriate standards
Why don't we read the manuals?
Usually it results from the lack of time and certainty about functionalities of purchased items. Unfortunately, networking and start-up of National Instruments SCXI -1125 alone are not enough. An instruction contains a number of clues concerning respective functionalities, safety rules, maintenance methods (what means should be used), eventual defects of National Instruments SCXI -1125, and methods of problem resolution. Eventually, when one still can't find the answer to his problems, he will be directed to the National Instruments service. Lately animated manuals and instructional videos are quite popular among customers. These kinds of user manuals are effective; they assure that a customer will familiarize himself with the whole material, and won't skip complicated, technical information of National Instruments SCXI -1125.
Why one should read the manuals?
It is mostly in the manuals where we will find the details concerning construction and possibility of the National Instruments SCXI -1125 item, and its use of respective accessory, as well as information concerning all the functions and facilities.
After a successful purchase of an item one should find a moment and get to know with every part of an instruction. Currently the manuals are carefully prearranged and translated, so they could be fully understood by its users. The manuals will serve as an informational aid.
Table of contents for the manual
CVI ™ , LabVIEW ™ , Natio nal Instru ments ™ ,n i . c o m ™ ,N I - D A Q ™ , and SCXI ™ [...]
SCXI-1125 Calibration Procedure 2 ni.com Equipment and Other T est Requirements This section describ[...]
© National Instruments Corpor ation 3 SCXI-1125 Calibration Procedur e Documentation The NI-DAQ Fun[...]
SCXI-1125 Calibration Procedure 4 ni.com Setting Up Y our Module Complete the following steps to set[...]
© National Instruments Corpor ation 5 SCXI-1125 Calibration Procedur e 7. Connect the calibrator to[...]
SCXI-1125 Calibration Procedure 6 ni.com 13. Call SCXI_Scale . Set the following parameters: • SCX[...]
© National Instruments Corpor ation 7 SCXI-1125 Calibration Procedur e • calOp — ND_SELF_CALIBR[...]
SCXI-1125 Calibration Procedure 8 ni.com • filterMode — 1 • cutoffFrequency — 10,000 9. Conn[...]
© National Instruments Corpor ation 9 SCXI-1125 Calibration Procedur e 16. Record binary1 with volt[...]
SCXI-1125 Calibration Procedure 10 ni.com V erifying Adjusted V alues After completing the adjustmen[...]
© National Instruments Corpor ation 11 SCXI-1125 Calibration Procedur e Front and Rear Signal Conne[...]
SCXI-1125 Calibration Procedure 12 ni.com Figure 2. Rear Signal Connector Pin Assignments SERCLK DIG[...]
© National Instruments Corpor ation 13 SCXI-1125 Calibration Procedur e Specifications The followin[...]
SCXI-1125 Calibration Procedure 14 ni.com 200 – 0.025 – 0.024750 – 0.02468642 – 0 .02481358 [...]