National Instruments SCC-LP04 manual
- Read online or download the manual
- 18 pages
- 0.18 mb
Go to page of
A good user manual
The rules should oblige the seller to give the purchaser an operating instrucion of National Instruments SCC-LP04, along with an item. The lack of an instruction or false information given to customer shall constitute grounds to apply for a complaint because of nonconformity of goods with the contract. In accordance with the law, a customer can receive an instruction in non-paper form; lately graphic and electronic forms of the manuals, as well as instructional videos have been majorly used. A necessary precondition for this is the unmistakable, legible character of an instruction.
What is an instruction?
The term originates from the Latin word „instructio”, which means organizing. Therefore, in an instruction of National Instruments SCC-LP04 one could find a process description. An instruction's purpose is to teach, to ease the start-up and an item's use or performance of certain activities. An instruction is a compilation of information about an item/a service, it is a clue.
Unfortunately, only a few customers devote their time to read an instruction of National Instruments SCC-LP04. A good user manual introduces us to a number of additional functionalities of the purchased item, and also helps us to avoid the formation of most of the defects.
What should a perfect user manual contain?
First and foremost, an user manual of National Instruments SCC-LP04 should contain:
- informations concerning technical data of National Instruments SCC-LP04
- name of the manufacturer and a year of construction of the National Instruments SCC-LP04 item
- rules of operation, control and maintenance of the National Instruments SCC-LP04 item
- safety signs and mark certificates which confirm compatibility with appropriate standards
Why don't we read the manuals?
Usually it results from the lack of time and certainty about functionalities of purchased items. Unfortunately, networking and start-up of National Instruments SCC-LP04 alone are not enough. An instruction contains a number of clues concerning respective functionalities, safety rules, maintenance methods (what means should be used), eventual defects of National Instruments SCC-LP04, and methods of problem resolution. Eventually, when one still can't find the answer to his problems, he will be directed to the National Instruments service. Lately animated manuals and instructional videos are quite popular among customers. These kinds of user manuals are effective; they assure that a customer will familiarize himself with the whole material, and won't skip complicated, technical information of National Instruments SCC-LP04.
Why one should read the manuals?
It is mostly in the manuals where we will find the details concerning construction and possibility of the National Instruments SCC-LP04 item, and its use of respective accessory, as well as information concerning all the functions and facilities.
After a successful purchase of an item one should find a moment and get to know with every part of an instruction. Currently the manuals are carefully prearranged and translated, so they could be fully understood by its users. The manuals will serve as an informational aid.
Table of contents for the manual
USER GUIDE SCC-LP Series Lowpass Filter Modules The SCC-LP Series lowpass filter mod ules contain fourth-order Butterworth filter circ uitry. They accept two di fferential input signals within a ± 10 V range. A differential ampl ifier attenuates each signal by a factor of two. The output of the amplifier passes through a fourth-order Butterworth f[...]
SCC-LP Series Lowpass Filter Module s User Guide 2 ni.com When symbol is marked on a product, it denotes a warning advising you to take precautions to av oid electrical shock. When symbol is marked on a product, it denotes a component that may be hot. T ouching this component may result in bodily injury . bold Bold text denotes items that you must [...]
© National Instruments Cor poration 3 SCC-LP Series Lowpass Filter Modul es User Guide What Y ou Need to Get Started To set up and use the SCC-LP XX , you need the following items: ❑ SC-2345/2350 with one o f the following: – SCC-PWR01 – SCC-PWR02 and the PS01 power supply – SCC-PWR03—requires a 7 to 42 VDC power supply (not included) ?[...]
SCC-LP Series Lowpass Filter Module s User Guide 4 ni.com Device Specific Information Note For general SCC module installation and signal connection information, and information about the SC-2350 carrier , refer to the SCC Quick Start Guide , av ailable for do wnload at ni.com/manuals . Installing the Module Caution Refer to the Read Me First: Safe[...]
© National Instruments Cor poration 5 SCC-LP Series Lowpass Filter Modul es User Guide The signal source can be floating or ground -referenced. The SCC-LP XX input circuitry includ es high-impedance bias resistors t ypically required for floating sources. Therefore, floati ng si gnal sources do not requir e external bias resistors connected to gro[...]
SCC-LP Series Lowpass Filter Module s User Guide 6 ni.com Using the SCC-LP XX when Scaling V oltage Measurements If you configur ed the SCC-LP XX u sing Measurement & Automation Explorer (MAX) and you are u sing NI -DAQ , the voltage reading you get from the E Series DAQ device accounts fo r the voltage scaling effect of the SCC-LP XX . O therw[...]
© National Instruments Cor poration 7 SCC-LP Series Lowpass Filter Modul es User Guide Input bias current .............. .............. ....... ± 2 nA typ ± 5 nA max 1 Nonlinearity ... ................. ................ ....... 0.004% of full scale Filter Characteristics Filter type ......... ................. ......... ........ .... Fourth-orde[...]
SCC-LP Series Lowpass Filter Module s User Guide 8 ni.com Stability Gain temperature coefficient .............. ....10 ppm/°C typ 20 ppm/°C m ax Offset-voltage temperature coefficient . ... .............. .........3.4 µ V/°C typ (RTI) 27 µ V/°C max Power Requirement Analog power SCC-LP01, SCC-LP02 ........... .........135 mW max +15 V .......[...]
© National Instruments Cor poration 9 SCC-LP Series Lowpass Filter Modul es User Guide Humidity ............. ................. .............. .... 10 to 90% RH, noncondensing Maximum altitude ................... ............... 2,000 m Pollution Degree (indoor use only) ........ 2 Safety The SCC-LP XX meets the requirements of the follo wing stan[...]
SCC-LP Series Lowpass Filter Module s User Guide 10 ni.com Theor y of Operation SCC-LP XX Performance The SCC-LP XX uses a Butterworth filter, which is characterized by maximal flatness in the passband with very sharp monotonic rol loff. It has a nonlinear phase response, the delay is not constant, and the st ep response exhibits a moderate amount [...]
© National Instruments Corpor ation 11 SCC-LP Series Lowpass Filter Modul es User Guide Figure 3. T ypical SCC-LP02 Response Curve Figure 4. T ypical SCC-LP03 Response Curve F requency (Hz) Amplitude (dB) – 25.0 1 2 3 4 5 10 20 25 50 100 – 20.0 – 15.0 – 10.0 – 5.0 0.0 F requency (Hz) Amplitude (dB) – 25.0 1 2 3 4 5 10 20 25 50 300 – [...]
SCC-LP Series Lowpass Filter Module s User Guide 12 ni.com Figure 5. T ypical SCC-LP04 Response Curve 1 2 3 4 5 10 20 50 1000 2000 –25.0 –20.0 –15.0 –10.0 –5.0 0.0 2.5 F requency (Hz) Amplitude (dB)[...]
© National Instruments Corpor ation 13 SCC-LP Series Lowpass Filter Modul es User Guide Figure 6 shows the theoretical transf er characteristics of the SCC-LP XX . Figure 6. Theoretical T r ansfer Characte ristics a. Frequency Response 0 0.1 0.15 0.5 Normalized F requency (f/fc) Amplitude (dB) 1 1.5 5 10 –20 –40 –60 –80 –100 b . Group De[...]
SCC-LP Series Lowpass Filter Module s User Guide 14 ni.com The horizontal axes of the first tw o plots are normali zed to the SCC-LP XX cutof f frequency . W hen the input freq uency (f ) equals the cuto ff fr equenc y (fc), the normalized frequency has a v alue of 1. The vertical axis of the third plot is normalized to the magnitude of the step in[...]
© National Instruments Corpor ation 15 SCC-LP Series Lowpass Filter Modul es User Guide Using the SCC-LP XX as an Antialiasing Filter Aliasing, a phenomenon of voltage-sampling systems, causes a high-fre quency signal to take on th e identity of a low-frequency signal. Figure 7. Aliasing of an Input Signal Freque ncy of 0.8 T imes the Sampling Rat[...]
SCC-LP Series Lowpass Filter Module s User Guide 16 ni.com Calibrating Gain Errors The SCC-LP XX is calibrated at the facto ry before shipment. If you want to calibrate the SCC-LP XX in your system, you need a vol tage source capable of providing a D C voltage up to ± 10 V that is several times more accu rate than the SCC itself. T o calibra te th[...]
© National Instruments Corpor ation 17 SCC-LP Series Lowpass Filter Modul es User Guide SCC-LP XX Module Pin Assignments Figure 8 shows the I/O connector pin s on the bottom of the modul e. Figure 8. SCC Module Bottom View 1P i n 1 2P i n 2 3P W B K e y 4 Pin 19 5P i n 2 0 5 4 2 1 3[...]
National Instruments™, NI™, ni.com ™, and NI-D AQ™ are trademarks of National Inst ruments Corporation. Product and com pany names mentioned herein are trademarks or trade names of their respective companies. Fo r patents covering National Instruments products, refer to the appropriate location: Help»Patents in you r software, th e patents[...]