National Instruments Portable Monitor Accessory NI PMA-1115 manual
- Read online or download the manual
- 15 pages
- 2.26 mb
Go to page of
A good user manual
The rules should oblige the seller to give the purchaser an operating instrucion of National Instruments Portable Monitor Accessory NI PMA-1115, along with an item. The lack of an instruction or false information given to customer shall constitute grounds to apply for a complaint because of nonconformity of goods with the contract. In accordance with the law, a customer can receive an instruction in non-paper form; lately graphic and electronic forms of the manuals, as well as instructional videos have been majorly used. A necessary precondition for this is the unmistakable, legible character of an instruction.
What is an instruction?
The term originates from the Latin word „instructio”, which means organizing. Therefore, in an instruction of National Instruments Portable Monitor Accessory NI PMA-1115 one could find a process description. An instruction's purpose is to teach, to ease the start-up and an item's use or performance of certain activities. An instruction is a compilation of information about an item/a service, it is a clue.
Unfortunately, only a few customers devote their time to read an instruction of National Instruments Portable Monitor Accessory NI PMA-1115. A good user manual introduces us to a number of additional functionalities of the purchased item, and also helps us to avoid the formation of most of the defects.
What should a perfect user manual contain?
First and foremost, an user manual of National Instruments Portable Monitor Accessory NI PMA-1115 should contain:
- informations concerning technical data of National Instruments Portable Monitor Accessory NI PMA-1115
- name of the manufacturer and a year of construction of the National Instruments Portable Monitor Accessory NI PMA-1115 item
- rules of operation, control and maintenance of the National Instruments Portable Monitor Accessory NI PMA-1115 item
- safety signs and mark certificates which confirm compatibility with appropriate standards
Why don't we read the manuals?
Usually it results from the lack of time and certainty about functionalities of purchased items. Unfortunately, networking and start-up of National Instruments Portable Monitor Accessory NI PMA-1115 alone are not enough. An instruction contains a number of clues concerning respective functionalities, safety rules, maintenance methods (what means should be used), eventual defects of National Instruments Portable Monitor Accessory NI PMA-1115, and methods of problem resolution. Eventually, when one still can't find the answer to his problems, he will be directed to the National Instruments service. Lately animated manuals and instructional videos are quite popular among customers. These kinds of user manuals are effective; they assure that a customer will familiarize himself with the whole material, and won't skip complicated, technical information of National Instruments Portable Monitor Accessory NI PMA-1115.
Why one should read the manuals?
It is mostly in the manuals where we will find the details concerning construction and possibility of the National Instruments Portable Monitor Accessory NI PMA-1115 item, and its use of respective accessory, as well as information concerning all the functions and facilities.
After a successful purchase of an item one should find a moment and get to know with every part of an instruction. Currently the manuals are carefully prearranged and translated, so they could be fully understood by its users. The manuals will serve as an informational aid.
Table of contents for the manual
USER GUIDE AND SPECIFICA TIONS NI PMA-1115 Portable Monitor Accessor y This document includes an over vie w and specif ications for the NI PMA-1115 Portable Monitor Accessory . Over view The National Instruments PMA-1115 is a monitor and keyboard accessory for use with N I 8-slot PXI chassis. It includes the following features: • 15 in. high-brig[...]
NI PMA-1115 User Guide and Specifications 2 ni.com • V ideo cabling to connect PMA-1115 to PXI controller • Right-angle-to-v ertical USB cable •D r i v e r C D • Cable ties to organize po wer cabling • Interlocking fastener to secure power adapter Mounting The NI PMA-1115 includes a complete mounting kit for attachmen t to any NI 8-Slot P[...]
© National Instruments Corporation 3 NI PMA-1115 User Guide and Specifications Figure 2. Removing the Handle 3. Remov e the handle from the chassi s. Retain the handle. Discard the screws. 4. Attach the adapter plate to the chassis using the supplied M4 × 8 pan-head screws, as shown in Figure 3. Figure 3. Installin g the Adapter Plate 1 Handle co[...]
NI PMA-1115 User Guide and Specifications 4 ni.com 5. Reinstall the handle using the su pplied 4-40 × 5/16 pan-head screws, as sho wn in Figur e 4. Reinstall the handle covers b y snapping them ov er the handle mounti ng plates. Figure 4. Reinstalling t he Hand le and Handle Covers 1 Handle covers 24 - 4 0 × 5/16 pan-head screws 1 2 (4x)[...]
© National Instruments Corporation 5 NI PMA-1115 User Guide and Specifications 6. Orient the chassis and PMA-1115 as shown in Figure 5. Install the remaining M4 × 8 pan-head scre ws. Figure 5. Installing the PMA-1115 7. Open the keyboard and LCD, as shown in Figure 6. Install the supplied 6-32 × 3/8 flat-head screws in the lo wer corners of the [...]
NI PMA-1115 User Guide and Specifications 6 ni.com 8. If desired, use the provided in terlocking fastener to attach the A C adapter to the unit. One sugges ted location is shown in Figure 7. Cable ties are also provided to organize the po wer cabling. Figure 7. Suggested Location for Attaching AC Adapter Note Do not place the A C adapter ov er any [...]
© National Instruments Corporation 7 NI PMA-1115 User Guide and Specifications Ordering Information To order the NI PMA-1115 kit or power cords, refer to the following kit numbers. NI PMA-1115 Kit NI PMA-1115 Portable Monitor Accessory ............ ............ ......... 780215-01 Additional Power Cords North American 240 VAC ... ........... .....[...]
NI PMA-1115 User Guide and Specifications 8 ni.com Display area ......... ....................... ............11.97 × 8.98 in. (30.41 × 22.81 cm) Contrast ratio ................... ................... .... 500:1 Brightness ....... ...................... ..................300 NTS Frame rate ..................... ...................... .... 60 Hz ([...]
© National Instruments Corporation 9 NI PMA-1115 User Guide and Specifications Operating Environment Ambient temperature range .................... 0 to 50 °C (Tested in accordance with IEC-60068-2-1 and IEC-60068-2-2. Meets MIL-PRF-28800F Class 3 low temperature lim it.) Relative humidity range ................ ......... 10 to 90%, noncondensing[...]
NI PMA-1115 User Guide and Specifications 10 ni.com Figure 8. T emperature/Humidity Relationship Shock and Vibration Operational shock ............... ....................10 g peak, half-sine, 11 ms pulse (Tested in accordance with IEC-60068-2-27.) Random Vibration Operating ...................... ...................... .... 5 to 500 Hz, 0.31 g rms[...]
© National Instruments Corporation 11 NI PMA-1115 User Guide and Specifications Safety and Compliance Safety This product is designed to meet the requirements of the follow ing standards of safety for information technolog y equipment: • IEC 60950-1, EN 60950-1 • UL 60950-1, CSA 60950-1 Note For UL and other safety certifications, refer to the[...]
NI PMA-1115 User Guide and Specifications 12 ni.com Environmental Management National Instruments i s committed to designing and manu facturing products in an environmental ly responsible mann er. NI recognizes that eliminating certa in hazardous substances from our products is beneficial not only to t he environment but also to NI customers. For a[...]
© National Instruments Corporation 13 NI PMA-1115 User Guide and Specifications Where to Go for Support The National Instruments Web site is your complete resource for technical support. At ni.com/support you have access to everything from troubleshooting and ap plication developmen t self-help resources to email and phone assistance from NI Appli[...]
NI PMA-1115 User Guide and Specifications 14 ni.com Important Information Warranty The PMA-1115 is warranted agains t defects in materials and workman ship for a period of one year from the date of shipment, as e videnced by receipts or other documentation. National Instruments will, at its option, repair or replace equi pment that proves to be def[...]