National Instruments NI USB-9472 manual
- Read online or download the manual
- 20 pages
- 0.67 mb
Go to page of
Similar user manuals
National Instruments Motion Axis Router MCA-77249 pages 0.09 mb
National Instruments PCI-232/2 Isolated19 pages 0.13 mb
National Instruments NI-DMM12 pages 0.11 mb
National Instruments VXI-MIO Series151 pages 0.81 mb
National Instruments DP840014 pages 0.25 mb
National Instruments Fieldbus41 pages 0.49 mb
National Instruments 8330 Series30 pages 0.48 mb
National Instruments NI-845x Software and Hardware6 pages 0.09 mb
A good user manual
The rules should oblige the seller to give the purchaser an operating instrucion of National Instruments NI USB-9472, along with an item. The lack of an instruction or false information given to customer shall constitute grounds to apply for a complaint because of nonconformity of goods with the contract. In accordance with the law, a customer can receive an instruction in non-paper form; lately graphic and electronic forms of the manuals, as well as instructional videos have been majorly used. A necessary precondition for this is the unmistakable, legible character of an instruction.
What is an instruction?
The term originates from the Latin word „instructio”, which means organizing. Therefore, in an instruction of National Instruments NI USB-9472 one could find a process description. An instruction's purpose is to teach, to ease the start-up and an item's use or performance of certain activities. An instruction is a compilation of information about an item/a service, it is a clue.
Unfortunately, only a few customers devote their time to read an instruction of National Instruments NI USB-9472. A good user manual introduces us to a number of additional functionalities of the purchased item, and also helps us to avoid the formation of most of the defects.
What should a perfect user manual contain?
First and foremost, an user manual of National Instruments NI USB-9472 should contain:
- informations concerning technical data of National Instruments NI USB-9472
- name of the manufacturer and a year of construction of the National Instruments NI USB-9472 item
- rules of operation, control and maintenance of the National Instruments NI USB-9472 item
- safety signs and mark certificates which confirm compatibility with appropriate standards
Why don't we read the manuals?
Usually it results from the lack of time and certainty about functionalities of purchased items. Unfortunately, networking and start-up of National Instruments NI USB-9472 alone are not enough. An instruction contains a number of clues concerning respective functionalities, safety rules, maintenance methods (what means should be used), eventual defects of National Instruments NI USB-9472, and methods of problem resolution. Eventually, when one still can't find the answer to his problems, he will be directed to the National Instruments service. Lately animated manuals and instructional videos are quite popular among customers. These kinds of user manuals are effective; they assure that a customer will familiarize himself with the whole material, and won't skip complicated, technical information of National Instruments NI USB-9472.
Why one should read the manuals?
It is mostly in the manuals where we will find the details concerning construction and possibility of the National Instruments NI USB-9472 item, and its use of respective accessory, as well as information concerning all the functions and facilities.
After a successful purchase of an item one should find a moment and get to know with every part of an instruction. Currently the manuals are carefully prearranged and translated, so they could be fully understood by its users. The manuals will serve as an informational aid.
Table of contents for the manual
USER GUIDE AND SPECIFICATIONS NI USB-9472 8-Channel, 24 V Logic, Sour cing Digital Output Device Thi[...]
NI USB-9472 U ser Guide and Specifications 2 ni.com Dimensions Figures 2 and 3 sho w the NI USB-9472[...]
© National Instruments Corporation 3 NI USB-9472 User Guid e and Specifications Safety Guidelines O[...]
NI USB-9472 U ser Guide and Specifications 4 ni.com Related Documentation Each application software [...]
© National Instruments Corporation 5 NI USB-9472 User Guid e and Specifications • T aking Measure[...]
NI USB-9472 U ser Guide and Specifications 6 ni.com ANSI C without NI Application Software The NI-D [...]
© National Instruments Corporation 7 NI USB-9472 User Guid e and Specifications T raining Courses I[...]
NI USB-9472 U ser Guide and Specifications 8 ni.com 3. Align the I/O module with the carrier , as sh[...]
© National Instruments Corporation 9 NI USB-9472 User Guid e and Specifications Mounting the NI USB[...]
NI USB-9472 U ser Guide and Specifications 10 ni.com Wiring the NI USB-9472 Device The NI USB-9472 p[...]
© National Instruments Corporation 11 NI USB-9472 User Gu id e and Specifications Y ou must connect[...]
NI USB-9472 U ser Guide and Specifications 12 ni.com Figure 7. Pin Assignments Connecting Devices to[...]
© National Instruments Corporation 13 NI USB-9472 User Gu id e and Specifications Figure 8. Connect[...]
NI USB-9472 U ser Guide and Specifications 14 ni.com Figure 10. Assembled High Voltage Screw T ermin[...]
© National Instruments Corporation 15 NI USB-9472 User Gu id e and Specifications Power Supplies an[...]
NI USB-9472 U ser Guide and Specifications 16 ni.com I/O protection Voltage . ........... ..........[...]
© National Instruments Corporation 17 NI USB-9472 User Gu id e and Specifications Weight With screw[...]
NI USB-9472 U ser Guide and Specifications 18 ni.com Safety Standards The NI USB-9472 product is des[...]
© National Instruments Corporation 19 NI USB-9472 User Gu id e and Specifications CE Compliance Thi[...]
National Instruments, NI, ni.com, and LabVIEW are tradem arks of National Instruments Corporation. R[...]