National Instruments NI USB-9472 manual
- Read online or download the manual
- 20 pages
- 0.67 mb
Go to page of
Similar user manuals
National Instruments GPIB-SCSI-A35 pages 0.98 mb
National Instruments NI USB-621x178 pages 3.87 mb
National Instruments NI 562038 pages 0.29 mb
National Instruments IMAQ Vision for LabWindows TM /CVI119 pages 2.11 mb
National Instruments NI 985227 pages 0.32 mb
National Instruments SCXI-1125101 pages 1.25 mb
National Instruments Relay Module SCC-RLY019 pages 0.26 mb
National Instruments SCXI-116160 pages 0.85 mb
A good user manual
The rules should oblige the seller to give the purchaser an operating instrucion of National Instruments NI USB-9472, along with an item. The lack of an instruction or false information given to customer shall constitute grounds to apply for a complaint because of nonconformity of goods with the contract. In accordance with the law, a customer can receive an instruction in non-paper form; lately graphic and electronic forms of the manuals, as well as instructional videos have been majorly used. A necessary precondition for this is the unmistakable, legible character of an instruction.
What is an instruction?
The term originates from the Latin word „instructio”, which means organizing. Therefore, in an instruction of National Instruments NI USB-9472 one could find a process description. An instruction's purpose is to teach, to ease the start-up and an item's use or performance of certain activities. An instruction is a compilation of information about an item/a service, it is a clue.
Unfortunately, only a few customers devote their time to read an instruction of National Instruments NI USB-9472. A good user manual introduces us to a number of additional functionalities of the purchased item, and also helps us to avoid the formation of most of the defects.
What should a perfect user manual contain?
First and foremost, an user manual of National Instruments NI USB-9472 should contain:
- informations concerning technical data of National Instruments NI USB-9472
- name of the manufacturer and a year of construction of the National Instruments NI USB-9472 item
- rules of operation, control and maintenance of the National Instruments NI USB-9472 item
- safety signs and mark certificates which confirm compatibility with appropriate standards
Why don't we read the manuals?
Usually it results from the lack of time and certainty about functionalities of purchased items. Unfortunately, networking and start-up of National Instruments NI USB-9472 alone are not enough. An instruction contains a number of clues concerning respective functionalities, safety rules, maintenance methods (what means should be used), eventual defects of National Instruments NI USB-9472, and methods of problem resolution. Eventually, when one still can't find the answer to his problems, he will be directed to the National Instruments service. Lately animated manuals and instructional videos are quite popular among customers. These kinds of user manuals are effective; they assure that a customer will familiarize himself with the whole material, and won't skip complicated, technical information of National Instruments NI USB-9472.
Why one should read the manuals?
It is mostly in the manuals where we will find the details concerning construction and possibility of the National Instruments NI USB-9472 item, and its use of respective accessory, as well as information concerning all the functions and facilities.
After a successful purchase of an item one should find a moment and get to know with every part of an instruction. Currently the manuals are carefully prearranged and translated, so they could be fully understood by its users. The manuals will serve as an informational aid.
Table of contents for the manual
USER GUIDE AND SPECIFICATIONS NI USB-9472 8-Channel, 24 V Logic, Sour cing Digital Output Device This user guide describes how to use the National Instruments NI USB-9472 and lists the device specif i cations. In this user guide, the NI USB-9472 with scr e w terminal and the NI USB-9472 with D-SUB are referred to inclusi vel y as the NI USB-9472. I[...]
NI USB-9472 U ser Guide and Specifications 2 ni.com Dimensions Figures 2 and 3 sho w the NI USB-9472 device dimensions. Figure 2. NI USB-9472 (Screw T erminal) Device in Millimeters (Inches) Figure 3. NI USB-9472 (D-SUB) Device in Millimeters (Inches) 25.34 (0.998) Hi-Speed USB Carrier NI USB-9162 140.23 mm (5.521 in.) 137.82 mm (5.426 in.) 120.68 [...]
© National Instruments Corporation 3 NI USB-9472 User Guid e and Specifications Safety Guidelines Operate the NI USB-9472 on ly as described in thes e operating instructions. Note Although the NI 9472 module may have more stringent certification standards than the NI USB-9472, when used with the NI USB-9162 carrier , the combined system m ay be li[...]
NI USB-9472 U ser Guide and Specifications 4 ni.com Related Documentation Each application software package an d driv er includes inform ation about writing applications for taking measur ements and controlling measurement devices. The follo wing references to documents assume you have NI-D A Qmx 8.7 or later, and where applicable, v ersion 7.1 or [...]
© National Instruments Corporation 5 NI USB-9472 User Guid e and Specifications • T aking Measurements —Contains the conceptual and how-to information you need to acquire and analyze measurement data in LabVIEW , including common measurements, measurement fundamentals, NI-DA Qmx key concepts, and device considerations. LabWindows/CVI The Data [...]
NI USB-9472 U ser Guide and Specifications 6 ni.com ANSI C without NI Application Software The NI-D A Qmx Help contains API o vervie ws and gen eral information about measurement concepts. Select Start»All Programs»National Instruments»NI-D A Q»NI-DA Qmx Help . The NI-D A Qm x C Refer ence Help describes the NI-D A Qmx Library functions, which [...]
© National Instruments Corporation 7 NI USB-9472 User Guid e and Specifications T raining Courses If you need more help getting started developing an appli cation with NI products, NI offers training courses. To enroll in a course or obtain a detailed course outline, r efer to ni.com/training . T echnical Support on the W eb For additional support[...]
NI USB-9472 U ser Guide and Specifications 8 ni.com 3. Align the I/O module with the carrier , as sho wn in Figure 4. Figure 4. Module Installation 4. Squeeze the latches and insert the NI 9472 module into the NI USB-9162 carrier . 5. Press f irmly on the connector side of the NI 9472 module un til the latches lock the module into place, as sho wn [...]
© National Instruments Corporation 9 NI USB-9472 User Guid e and Specifications Mounting the NI USB-9472 to a Panel Threaded inserts are located in the NI USB-9472 for mounting it to a panel. Refer to Figure 6 for dimensions. Figur e 6. Module Dimensions in Millimeters (Inches) Connecting the NI USB-9472 to a Computer Plug one end of the USB cable[...]
NI USB-9472 U ser Guide and Specifications 10 ni.com Wiring the NI USB-9472 Device The NI USB-9472 provides connections for eigh t digital output channel s. Caution A high v oltage screw terminal backshell must be installed when using hazardous voltages (>42.4 V pk , 60 VDC). Note T able 2 illustrates the accessories a v ailable from ni. com for[...]
© National Instruments Corporation 11 NI USB-9472 User Gu id e and Specifications Y ou must connect an external po wer supply to the NI USB-9472. This power supply provides the current for the de vices you connect to the module. Connect the positi ve lead of the po wer supply to the supply terminal or pin, V sup , and the negati ve lead of the po [...]
NI USB-9472 U ser Guide and Specifications 12 ni.com Figure 7. Pin Assignments Connecting Devices to the NI USB-9472 The NI USB-9472 has current sourcing outputs , meaning the DO term inal or pin drives current or applies vo ltage to the device to which it is connected. Y ou can connect sinking-input devices, such as NPN, open collectors, normally [...]
© National Instruments Corporation 13 NI USB-9472 User Gu id e and Specifications Figure 8. Connecting a Device to the NI USB-9472 When the channel is on, the channel LED is on and the terminal or pin associated with the chan nel driv es a current or applies a v oltage to the device . When the channel is of f, the ch annel LED is of f and the term[...]
NI USB-9472 U ser Guide and Specifications 14 ni.com Figure 10. Assembled High Voltage Screw T erminal Back shell I/O Protection The NI USB-9472 is s hort-circuit-proof in acco rdance with IEC 1131-2 and provides overcurrent protection. Understanding Short-Ci rcuit-Proof Devices Each channel on the NI USB-9472 has circ uitry that protects it from c[...]
© National Instruments Corporation 15 NI USB-9472 User Gu id e and Specifications Power Supplies and Overcurrent Conditions If a short-circuit occu rs, the current through DO can exceed the current rating for the power supply and the m aximum continuous ou tput current for the NI USB-9472. If the po wer supply you are using with the NI USB-9472 ca[...]
NI USB-9472 U ser Guide and Specifications 16 ni.com I/O protection Voltage . ........... .............. .............. ....30 V max Reversed voltage ................. ............None Trip currents Minimum possible trip current ... .............. .............. .6 A Minimum guaranteed trip current ... .............. .............. .13 A Trip time [...]
© National Instruments Corporation 17 NI USB-9472 User Gu id e and Specifications Weight With screw terminal ............... ......... Approx. 250 g (8.8 oz) With D-SUB ............ .............. ......... Approx. 245 g (8.6 oz) Safety If you need to clean the module, wipe it with a dry towel. Safety Vo ltages NI USB-9472 with Screw T erminal Cha[...]
NI USB-9472 U ser Guide and Specifications 18 ni.com Safety Standards The NI USB-9472 product is designed to meet the requirements of th e following standards of saf ety for el ectrical equipment for measurement, control, and laboratory use: • IEC 61010-1, EN-61010-1 • UL 61010-1, CSA 61010-1 Note For UL and other safety certifications, refer t[...]
© National Instruments Corporation 19 NI USB-9472 User Gu id e and Specifications CE Compliance This product meets the essential re quirements of applicable European Directives, as amended fo r CE marking, as follows: • 2006/95/EC; Lo w-V oltage Directi ve (safety) • 2004/108/EC; Elect romagnetic Compatibility Directiv e (EMC) Note Refer to th[...]