National Instruments 7602 manual
- Read online or download the manual
- 31 pages
- 0.49 mb
Go to page of
A good user manual
The rules should oblige the seller to give the purchaser an operating instrucion of National Instruments 7602, along with an item. The lack of an instruction or false information given to customer shall constitute grounds to apply for a complaint because of nonconformity of goods with the contract. In accordance with the law, a customer can receive an instruction in non-paper form; lately graphic and electronic forms of the manuals, as well as instructional videos have been majorly used. A necessary precondition for this is the unmistakable, legible character of an instruction.
What is an instruction?
The term originates from the Latin word „instructio”, which means organizing. Therefore, in an instruction of National Instruments 7602 one could find a process description. An instruction's purpose is to teach, to ease the start-up and an item's use or performance of certain activities. An instruction is a compilation of information about an item/a service, it is a clue.
Unfortunately, only a few customers devote their time to read an instruction of National Instruments 7602. A good user manual introduces us to a number of additional functionalities of the purchased item, and also helps us to avoid the formation of most of the defects.
What should a perfect user manual contain?
First and foremost, an user manual of National Instruments 7602 should contain:
- informations concerning technical data of National Instruments 7602
- name of the manufacturer and a year of construction of the National Instruments 7602 item
- rules of operation, control and maintenance of the National Instruments 7602 item
- safety signs and mark certificates which confirm compatibility with appropriate standards
Why don't we read the manuals?
Usually it results from the lack of time and certainty about functionalities of purchased items. Unfortunately, networking and start-up of National Instruments 7602 alone are not enough. An instruction contains a number of clues concerning respective functionalities, safety rules, maintenance methods (what means should be used), eventual defects of National Instruments 7602, and methods of problem resolution. Eventually, when one still can't find the answer to his problems, he will be directed to the National Instruments service. Lately animated manuals and instructional videos are quite popular among customers. These kinds of user manuals are effective; they assure that a customer will familiarize himself with the whole material, and won't skip complicated, technical information of National Instruments 7602.
Why one should read the manuals?
It is mostly in the manuals where we will find the details concerning construction and possibility of the National Instruments 7602 item, and its use of respective accessory, as well as information concerning all the functions and facilities.
After a successful purchase of an item one should find a moment and get to know with every part of an instruction. Currently the manuals are carefully prearranged and translated, so they could be fully understood by its users. The manuals will serve as an informational aid.
Table of contents for the manual
National Instruments™, NI™, and ni.com™ are t rademarks of National Instruments Corporation. P[...]
MID-7604/7602 Power Drive 2 ni.com Technical Support Resources ............. ............... .......[...]
© National Instruments Corporation 3 MID-7604/7602 Power Drive • Connect the equipment into an ou[...]
MID-7604/7602 Power Drive 4 ni.com Introduction National Instruments ’ MID-7 604/7602 power dri ve[...]
© National Instruments Corporation 5 MID-7604/7602 Power Drive ❑ One of the follo wing shielded c[...]
MID-7604/7602 Power Drive 6 ni.com • Pollution D egree 2 — Normally only noncon ductiv e pollu t[...]
© National Instruments Corporation 7 MID-7604/7602 Power Drive The following is a diagram of a samp[...]
MID-7604/7602 Power Drive 8 ni.com The two rock er switches on the MID-7604/7602 front panel are the[...]
© National Instruments Corporation 9 MID-7604/7602 Power Drive Caution Be sure to turn off the EN A[...]
MID-7604/7602 Power Drive 10 ni.com If the DC power supplies are acti ve, the green po wer LED illum[...]
© National Instruments Corporation 11 MID-7604/7602 Power Drive Front Panel DIP Switch Settings The[...]
MID-7604/7602 Power Drive 12 ni.com Microstepping Selection The MID-7604/7602 uses bipolar chopper, [...]
© National Instruments Corporation 13 MID-7604/7602 Power Drive Output Current Settings The MID-760[...]
MID-7604/7602 Power Drive 14 ni.com T o determine the maximum total power dissipation of all of the [...]
© National Instruments Corporation 15 MID-7604/7602 Power Drive Inhibit Input Polarity Setting The [...]
MID-7604/7602 Power Drive 16 ni.com Back Panel Connector Wiring Motor Power T erminal Blocks Each ax[...]
© National Instruments Corporation 17 MID-7604/7602 Power Drive Caution Ne ver connect unused cent [...]
MID-7604/7602 Power Drive 18 ni.com If the encoder cable length is great er than 10 ft, use encoders[...]
© National Instruments Corporation 19 MID-7604/7602 Power Drive Figure 8 shows clockwise and counte[...]
MID-7604/7602 Power Drive 20 ni.com Limit Switch T erminal Blocks For end-of-tra vel limit, home, in[...]
© National Instruments Corporation 21 MID-7604/7602 Power Drive Figure 11. T rigger T erminal Block[...]
MID-7604/7602 Power Drive 22 ni.com Accessories for Optional Use Strain-Relief Bar The strain-relief[...]
© National Instruments Corporation 23 MID-7604/7602 Power Drive Panel-Mount Kit (Included) The pane[...]
MID-7604/7602 Power Drive 24 ni.com Stepper Motor Configurations This section describes the various [...]
© National Instruments Corporation 25 MID-7604/7602 Power Drive For maximum fle xibility , y ou can[...]
MID-7604/7602 Power Drive 26 ni.com Notice that an 8-wire motor wired in series is virtually identic[...]
© National Instruments Corporation 27 MID-7604/7602 Power Drive Specifications The follo wing speci[...]
MID-7604/7602 Power Drive 28 ni.com Inhibit Inputs Voltage range ........................ ..........[...]
© National Instruments Corporation 29 MID-7604/7602 Power Drive Breakpoints ............ ..........[...]
MID-7604/7602 Power Drive 30 ni.com Environment Operating temperature ................... ......... [...]
© National Instruments Corporation 31 MID-7604/7602 Power Drive T echnical Support Resources NI We [...]